Simultaneously evaluating postthaw viability and acrosome integrity of spermatozoa by flow cytometry would provide a valuable testing tool in both research and routine work. In the present study, a new triple-stain combination was developed for the simultaneous evaluation of viability and acrosome integrity of bovine sperm processed in egg yolk-based extender by flow cytometer. SYBR-14 and propidium iodide (PI) enabled the discrimination of sperm cells from egg yolk and debris particles, which was instrumental for the flow cytometric analyses of frozen-thawed bovine sperm, because it implied that washing steps to remove egg yolk were no longer required. In addition, phycoerythrin-conjugated peanut agglutinin (PE-PNA) was used to discriminate acrosome-damaged/reacted sperm cells from acrosome-intact cells. Repeatability was calculated using two processed ejaculates of 10 bulls. Three straws per batch were analyzed in duplicate measurements. Method-agreement analysis between the SYBR-14/PE-PNA/PI and fluorescein isothiocyanate (FITC)-conjugated PNA was performed, with FITC-PNA/PI staining being carried out on 14 frozen-thawed semen samples immediately after thawing and after a 3-h incubation at 37°C. The British Standards Institution repeatability index of the SYBR-14/PE-PNA/PI combination was 2.6%. On average, the FITC-PNA/PI method showed a 6.3% overestimation of the live and acrosome-intact sperm cell subpopulation. In conclusion, the new triple-stain combination is highly repeatable and easy to use in routine application, and it provides a more precise estimate for the rate of sperm cells with intact head membrane and acrosome compared to the generally used and validated FITC-PNA/PI staining.
You have requested a machine translation of selected content from our databases. This functionality is provided solely for your convenience and is in no way intended to replace human translation. Neither BioOne nor the owners and publishers of the content make, and they explicitly disclaim, any express or implied representations or warranties of any kind, including, without limitation, representations and warranties as to the functionality of the translation feature or the accuracy or completeness of the translations.
Translations are not retained in our system. Your use of this feature and the translations is subject to all use restrictions contained in the Terms and Conditions of Use of the BioOne website.