The laryngeal mask airway is an alternative to endotracheal intubation that achieves control of the airway by creating a seal around the larynx with an inflatable cuff. This study compared use of the laryngeal mask airway with endotracheal intubation in anesthetized western lowland gorillas (Gorilla gorilla gorilla). Eight adult gorillas were immobilized for routine and diagnostic purposes for a total of nine anesthetic events. During each anesthetic event, gorillas were either intubated (n = 4; group A) or fitted with a laryngeal mask airway (n = 5; group B). Time required to place each airway device, physiologic parameters, and arterial blood gas were measured and compared between the two groups. There were no significant differences between the two groups for time required to place airway device, heart rate, hemoglobin oxygen saturation, end-tidal carbon dioxide, arterial partial pressure of carbon dioxide, or arterial pH between the two groups. Mean arterial partial pressure of oxygen was significantly greater in group B, 15 (group A: 94 ± 44 mm Hg; group B: 408 ± 36 mm Hg; P = 0.0025) and 45 (group A: 104 ± 21 mm Hg; group B: 407 ± 77 mm Hg; P = 0.0026) min after airway device placement. Mean respiratory rate was significantly greater in group A at multiple time points. Mean arterial pressure (group A: 129 ± 16 mm Hg; group B: 60 ± 8 mm Hg) and diastolic blood pressure (group A: 115 ± 21 mm Hg; group B: 36 ± 10 mm Hg) were significantly greater in group A at the time of airway device placement. The laryngeal mask airway maintained oxygenation and ventilation effectively in all gorillas and is a useful alternative to endotracheal intubation in western lowland gorillas.
You have requested a machine translation of selected content from our databases. This functionality is provided solely for your convenience and is in no way intended to replace human translation. Neither BioOne nor the owners and publishers of the content make, and they explicitly disclaim, any express or implied representations or warranties of any kind, including, without limitation, representations and warranties as to the functionality of the translation feature or the accuracy or completeness of the translations.
Translations are not retained in our system. Your use of this feature and the translations is subject to all use restrictions contained in the Terms and Conditions of Use of the BioOne website.