Blood pressure measurement reveals important insights into the health of conscious and anesthetized individuals. This is of particular interest in cheetahs (Acinonyx jubatus), which in captivity are known to suffer from chronic diseases that may be associated with hypertension and which often require immobilization for transport or veterinary treatment. Invasive testing methods are considered the gold standard but are not practical in many settings. Consequently, it is important to evaluate the use of noninvasive methods in this species. Measurements for systolic, diastolic, and mean arterial pressure obtained using high-definition oscillometry (HDO) at the coccygeal artery were compared to simultaneous direct measurements obtained via catheterization of the femoral or dorsal pedal artery in eight anesthetized captive cheetahs during nine anesthetic events. Overall, HDO and direct measurements agreed most closely for mean arterial pressure, and the poorest agreement was observed for systolic pressure. There was a tendency for low diastolic pressures to be underestimated and for high diastolic pressures to be overestimated. Across all three parameters, HDO measurements from the tail overestimated directly measured pressures in the femoral artery and underestimated those in the dorsal pedal artery. HDO agreed most closely with directly measured dorsal pedal pressures. Mean arterial pressure showed the greatest precision (standard deviation of 10.2 mm Hg) and lowest bias (−1.2 mm Hg), with 75.9% of readings within 10 mm Hg of the direct dorsal pedal pressure. Agreement with systolic pressure was hindered by a high bias (−10.4 mm Hg), but if a correction factor of 10 mm Hg was applied to all systolic measurements, agreement was improved and 65.7% of readings were within 10 mm Hg of the direct pressure. When compared to criteria defined by the American College of Veterinary Internal Medicine for validation of blood pressure devices, results were favorable, but a limited sample size prevented formal validation.
You have requested a machine translation of selected content from our databases. This functionality is provided solely for your convenience and is in no way intended to replace human translation. Neither BioOne nor the owners and publishers of the content make, and they explicitly disclaim, any express or implied representations or warranties of any kind, including, without limitation, representations and warranties as to the functionality of the translation feature or the accuracy or completeness of the translations.
Translations are not retained in our system. Your use of this feature and the translations is subject to all use restrictions contained in the Terms and Conditions of Use of the BioOne website.